Payment & Shipping Terms:
|Measurement Technology:||Eddy Current||Operating Frequency:||60KHz,240KHz|
|Display Screen:||240X320 Pixels TFT-LCD||Measuring Range:||6.9-110%IACS(4 -64MS/m)|
|Distinguishing Rate:||Conductivity:0.01% IACS(MS/m)||Weight:||245g|
eddy current testing machine,
eddy current tester
Electrical Eddy Current Testing Equipment Digital Portable 60KHz
HEC-102 is a kind of eddy current electrical conductivity meter, which is designed for quick & convenient measurement of material property, such as material separate,quality control,material state check and so on. It uses the electromagnetism testing principle. Testing objects focus on non-ferromagnetic material.
Eddy current exam is an NDT method established on the basis of electromagnetic induction principle; it is suitable for the conductive materials. Put the conductor in the alternating magnetic field, if inductive current exists, namely eddy current. On account of the change of each conductor’s own factors (for example: conductivity, magnetic conductivity, shape, size and flaw, etc), will cause the change of the current, so we can use this phenomenon to judge the conductor’s character or state and whether flaw exists.
Eddy current test is to put the conductor close to the alternating current coil, the coil establishes alternating magnetic field, this alternating magnetic field transits the conductor, and has the electromagnetic induction with it, establishes eddy current in the conductor. The current in the conductor also can produce its own magnetic field and can change the alternating magnetic field to be strong or weak, then causes the change of the voltage and R resistance. Therefore if the conductor’s surface or the near surface has flaw or the metal material changes, the intensity and the distribution of the eddy current will also change, so as the change of the voltage and R resistance, according to this change, the eddy current may indirectly know about the flaw in the conductor and whether the performance of the metal material changed.
★The meter uses 60 KHz (aviation industry standard) to inspirit, and the testing data can be read in two kinds of unit: %IACS or MS/m.
★Its big typeface, the back light illumination designs are advantageous for users to take the testing data even in low light condition.
★Two kinds of operate languages satisfy different national demand.
★It uses the high property battery to make sure it keeps more running time, and because of its mini-size, it is easy to carry and to grasp hold.
★The design of the meter is more advantages: user can replace the probe in the outdoor, don’t need to return to the company to adjust the probe into matching the meter.
★It can store 1000 measurement data.
|Measurement technology||Eddy current|
|Display screen||240X320 pixels TFT-LCD|
|Instrument case||Anti-intense impact, water-proofing polyester; Toolbox.|
|conductivity standard block||Verify and guarantee temperature compensation.|
|Resistivity||Correspond the Conductivity|
|Distinguishing rate|| |
|Measuring accuracy|| |
0°C to 50°C
|Temperature compensation||Automatic compensation to the value of 20 ℃.|
|Normal work environment||Relative humidity||0～95%|
|Data storage||Can store 1000 measurement data.|
Applicable to minimum measuring area diameter at 60KHz is 12mm.
Applicable to minimum measuring area diameter at 60KHz is 8mm.
Note:The conductivity measurement result automatically rectifies for 20°
★ Test the conductivity of aluminum, copper and other non-ferrous magnetic metal in the processing industry.
★ In the aerospace and automotive industries,monitor the process of heat treatment, the strength and hardness of aluminum alloy.
★ Test the conductivity aluminum when it has not been oxidized.
★ Test the purity grade of materials.
★ Test materials resistivity.
★ Material Thermal Performance analysis.
Hand-held eddy current conductivity tester for NF-metals with automatic normalization of the conductivity values to 20°C through temperature sensor.
v. Products use and maintain
1. In order to prevent electric shock or catch fire, please use the suitable power adapter
2. Avoid to drop and hit the instrument.
3. Please keep the instrument clean and dry, please check the probe and standard block weather are damaged or corrupted before using.
4. Please do not operate in wet, explosive, chemic or high temperature environment.
5. Do not voluntarily replace the spare part, if the product is abnormal or has suspicious breakdown, please contact with technology department or serviceman of our company for help.
Tel: 0086 139 109 83110