Payment & Shipping Terms:
|Material:||ABS||Application:||Measurement Of Light Sources And Irradiation Systems, Transmission Measurements, UV Radiation Safety|
|UV－365 Probe:||λ：(320～400)nm；λP＝365nm||UV－420 Probe:||λ：(375～475)nm；λP＝420nm|
|Measuring Range:||(0.1～199.9×103)μW/cm2 Parasitic Light：UV365：less Than 0.02％ UV420：less Than 0.02％||Accuracy:||±10％|
|Size And Weight:||180mm×80mm×36mm；0.2kg||Power:||9V Battery（6F22）1pc|
magnetic flaw detector,
mt test specimen
UV Radiometer UV-A Ultraviolet Irradiance Meter UV Light, Ultraviolet Illuminometer
Aplplication: Measuremen of light sources and irradiation systems, transmission measurements, UV radiation safety.
The instrument apply to different ultraviolet irradiation measurement field, such as photochemical, polymer materials aging, flaw detection, UV light source, plant cultivation and large scale integrated circuit
Pparasitic light:UV365:less than 0.02％
UV420:less than 0.02％
Cosine:According to China and international standard
Response time:1 second
Working environment:Temperature: (0~40)℃；Humidity＜85％RH
Size and weight:180mm×80mm×36mm；0.2kg
The instrument has two parts, UV365(or UV420) and reading unit. The two parts can be connected by cable using plug and socket.
The function of each key on the left as follows,
Power: connected state after pressing; off-state after raising this button.
Hold: data keep state after pressing, data sampling state after raising this button.(please raising this button when do measurement)
UV365 please pressing this button when uses 365 probe(raising UV420 button)
UV420 please pressing this button when uses 365 probe(raising UV420 button)
“×10” ranges buttons
Then connect the 365 probe(or 420 probe) to the socket of the reading unit. Open the probe cover and put the photosurface of probe on the measure position. So the product of digital displayed on widow and range factor is irradiance value (unit: μW/cm2).
If want to keep the data, you can press “hold” button ( note, please press the “hold” button after raising range buttons). Raise the “hold” button after reading the value, recovering to sampling state.
Cut off the power button after measuring.
Please store the instrument in a clean environment, temperature (0～40)℃ and humidity <85%RH to avoid any damage.
We can repair or change it during one year freely if there is any problems which is not caused by improper use and maintain.
Tel: 0086 139 109 83110