Payment & Shipping Terms:
|Radiation Angle:||40±5°||Usage:||Industrial X-Ray Testing,weld Testing,Industrial X Ray Testing Machine|
|Power Supply:||AC220V±10%,50HZ||Application:||Industrial X-Ray Testing,welding Seam Detecting|
|Warranty:||12 Months||Exposure Time:||5 Minutes|
|Controller Size:||350*290*170mm||Generator Size:||335×335×700mm|
welding x ray machine,
industrial x ray machine,
x ray inspection equipment
320KV Ceramic tube penetrant 40mm Directional Radiographic inspection machine
Main technical parameters
|Max Penetration thickness Steel (A3)||55mm|
|Film Type||Agfa D7|
Double lead foil Intensifying screen
|Input||Voltage (V)||220V±10% 50―60Hz|
|Output||Tube voltage (KVP)||170-320Kv|
|Tube current (mA)||5|
|Operation Mode||Intermittent operation, Radiating time to cooling time is 1:1|
|Circuit Structure||Micro-computer control ,Modular structure,Voice prompt|
|Cooling||Force air cooling|
|Insulation||SF6 Gas Insulation|
|Directional glass X-ray tube||1 Pcs|
|High voltage joins cable||1 Pcs||25 Meters|
|Power cable||1 Pcs||10 Meters|
|Grounding cable||1 Pcs||5 Meters|
|(Including: alarm lamp,1 Pcs; lead letter, 1 box; Film dark cassettes 5 pcs; intensifying screen 5 pcs; Wire penetrameter (IQI) 1 set; fuse1 pc)|
1.Check and count all goods to be received per the packing list. And then check the quality of the parts for damage,lost and mistake.
2.Keep the controller in a flat,dry and dustless place, switch off the power and connect the power cable.
The power supply of the unit is single phase 220V±10% 50-60HZ)
3.If necessary,one end of earth wire can be connected to the earth terminal of the controller and the other end to the special grounding wire.
4.Connect controller to x-ray generator by special 9 cores cable,and form better contact.
5.Check the gas pressure gauge on x-ray generator and make sure it range from 0.35Mpa to 0.45Mpa.If not,the unit is not allowed to turn on so as to avoid any damage to the x-ray generator.
This machine is composed of intelligent controller-ray generator, power cable and connecting cable.
1.The controller will consist of the following sections:
Front Panel: LED display, status and failure indicator, touch keyboards (see the introduction of control panel)
Side Panel: power cable socket and connecting cable socket, power switch,key switch, fuse socket, warning lamp and wire connecting terminal of door-machine interlock.
Internal Circuit: power circuit, rectifying circuit, filter circuit, inverter circuit, control circuit and other circuited.
2.The x-ray generator will consist of the following sections:
X-ray tube, High voltage transformer, Temperature relay, Gas pressure gauge and Cooling fan etc.
3.The schematic diagram of the complete machine:
Testing platform, IQI/wire penetrameter, lead letter/digits, lead tape, Cassette, lead screen, personal dosimeter, film viewer, densitometer, film developer, dark room light, RT calibration block etc.
Tel: 0086 139 109 83110